|Country:||Antigua & Barbuda|
|Published (Last):||8 March 2004|
|PDF File Size:||4.79 Mb|
|ePub File Size:||16.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Download this manual. Table of Contents. Page 4: Safety Warnings 1. Therefore, read through these operating instructions before using the instrument. Otherwise, the use of the instrument may cause sparking, which can lead to an explosion. Remove the batteries if the instrument is to be stored and will not be in use for a long period. Page 7: Features 2. Page 8: Specifications 3.
Page 9 Current Consumption: 80mA approx. Threshold voltages between Green and Yellow, and Yellow and Red are approximately 9. Page 11 KEWA Page Instrument Layout 4. Page Preparations For Tests 5. If not, adjust it by rotating the Meter Movement Zero Adjust with a small screwdriver. If the Battery Alarm is not lit or the digital display does not operate, confirm that the batteries are installed correctly and charge the batteries.
Page Operating Instructions 6. Use a high voltage detector and verify there is no voltage presence before making a measurement. Do not touch exposed metal pipes, outlets, fixtures, etc. Keep you isolated from ground by using dry clothing, rubber shoes, rubber mats, or any approved insulating material. Page 19 How to use the Guard Terminal In cable insulation tests, wind a conductive wire around insulation of the cable under test and connect it to the Guard terminal with the Guard lead as per Fig.
This is to move out the surface leakage resistance of the cable insulation to make the test results accurate. Page Battery Charging 7. To avoid a risk of electrical shock, do not connect the test leads to the equipment under test and set the Function switch to OFF when replacing the batteries. Page 21 2 If the instrument does not turn operative with the Function switch set to the H. SET position, charge the batteries according to section Center negative and center positive; both are useable.
Page Battery Refresh 8. But their capacity may fairly reduce before their lives wear out. You can refresh them in the following steps. Set the Function switch to H. Do not press the Test button. Page Battery Replacement 9. Page Battery Handling Precautions Page Cleaning Of Meter Cover But in the dry time of winter static electricity sometimes builds upon the meter cover due to the characteristic of plastic.
When the pointer deflects by touching the surface of this tester or zero adjustment cannot be made, do not try to make measurement. Print page 1 Print document 28 pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password.
Upload from disk. Upload from URL.
Camberwell, Canterbury, Deepdene, Hawthorn East & 2 others – Rooms for Rent
Quick Links. Download this manual. Table of Contents. Page 4: Safety Warnings 1.
Japan kyoritsu kew 3125a model 3124 model 3125 insulation resistance tester 3124 s
KEW - 3124 Analog High Voltage Checker