Tell us about your needs: Comments. Alltest Instruments. Sign up to receive product inventory and pricing updates! Estimated Rental Rate : Get Quote.
|Published (Last):||24 December 2015|
|PDF File Size:||16.66 Mb|
|ePub File Size:||16.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Go to Solution. Have you consulted the Tek manual to see what these Tek-generated errors are? It would be helpful to see the whole error message window, not just the left part of it. Use "Configure Record Data. For more details have a look at appendix B of the programmer's manual the composition of waveform data paragraph. You should then mark the post that helped you as the accepted solution. You should also post the modified vi that works, this could help other people with with a similar problem find a solution.
This average function seems having some bugs. When I changed the record length of data by using 'configure timebase. When the record legnth of data doesn't change.
It is correct. United States. Turn on suggestions. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Showing results for. Search instead for. Did you mean:. Tektronix DPO average function Solved! Tektronix DPO average function. Does anyone have the same problem before? Virus scan in progress. Please wait to download attachments. Message 1 of 8. Re: Tektronix DPO average function. Proven Zealot. Bill Mid-Level minion. My support system ensures that I don't look totally incompetent. Proud to say that I've progressed beyond knowing just enough to be dangerous. I now know enough to know that I have no clue about anything at all.
Message 2 of 8. Trusted Enthusiast. Ben64 The best way to say thanks is to give kudos! Message 3 of 8. My bad. Message 4 of 8. Thanks for your information. Message 5 of 8. Accepted by topic author Willy Message 6 of 8. Here is the solution to use average function in Tektronix DPO Message 7 of 8.
Any ideas? Thank you very much! Message 8 of 8.
Tektronix DPO2024 Digital Phosphor Oscilloscope
Debug faster and easier than ever before with Wave Inspector's unique ability to zoom, pan, play, pause and set marks — helping you to efficiently navigate through long records. Need digital channels? The MSO Series adds 16 integrated digital channels, enabling you to visualize and time correlate analog and digital signals on a single instrument. This integration extends triggering functionality across all 20 channels — which is ideal for debugging mixed analog and digital designs.